T4020-01 Gauge configuration Kit
This kit allows any user to connect to the T5020 and Z5020 Gauge range using a lap-top.
The free configuration software included enables
Tank shape set, SG of product set,, RS485 node ID to be set, Sensor parameters to be set, Alarm trigger percentages and direction to be set.
All possible settings that are shown in the Documentation section showing Modbus details.
This connection lead has a USB 2 connection to Lap-top/PC
This lead will self download the Drivers needed in later models. In Earlier versions you will need to download the Drivers. (See Document Download Below)
Connect the 4 way green connector to the T4020 / T5020 board. (There is only one green socket so this can not be confused).
version 220.127.116.11 will operate with Windows XP, 7, 8.
version 18.104.22.168 will operate with windows 8 and 10
See Documentation section for set up and operation.
Used for Calibrating all T4020 and T5020 and Z5020 Tank Gauges
Q1/ The Laptop will not communicate with the Gauge.
A/ Check that the "Port" is the correct one in the Configurator "FILE" "Program Settings"
In most cases the reason why the PC can not communicate is because either the DRIVERS have not been installed. (from 2016 this is an automated download, so the PC must be connected to the internet to download the drivers.)
Then find the right Port assignment in device hardware profiles
Then it should work. We are not "I.T." specialists so please do not call us if you can not communicate with the gauge, please seek IT help.
B/ If still not talking, check "Modbus and Standard" settings, and NODE ID. (See How to set up the Copnfigurator in the Document Download section above, and also I & O Manual for Gauges for this detail).
C/ If the lights are not flashing on the USB Lead, this is a Driver issue.
D/ If the configurator continually drops out, it may be that you are using an older version than the guage would like. Keep in mind 22.214.171.124 is newer than 0.0.0.17 ! Latest is 126.96.36.199